Generated Mon, 17 Oct 2016 09:05:24 GMT by s_ac15 (squid/3.5.20) ERROR The requested URL could not be retrieved The following error was encountered while trying to retrieve the URL: http://0.0.0.10/ Connection If you experience this error, check the following: Verify the command you sent is valid for that instrument. This error occurs when the I/O operation is aborted due to timeout, ibstop, or Device Clear. 7 Nonexistent GPIB interface. For example, the default interface name for NI boards is GPIB0, but you may misspell it as GPIBO (with an "oh" instead of a zero). http://imagextension.com/error-code/gpib-read-error-code-6.php
Answered Your Question? For example, the "*IDN?" message from the previous example is only understood by IEEE 488.2 compliant instruments. Please Contact NI for all product and support inquiries. Send an EOS (End of String) character at the end (carriage return('\r') or a linefeed('\n')). http://digital.ni.com/public.nsf/allkb/2FA525A8585A92E9862566EE002A3755
This error occurs when the board is not the System Controller but needs to be to perform the requested operation. 6 I/O operation aborted. This error occurs when the board is the Controller-In-Charge and is not properly addressed before starting a transfer. Code Description 0 Error connecting to driver or device. For example, if you disabled hardware DMA by removing the DRQ/DACK jumpers on her legacy AT-GPIB/TNT, a call to ibconfigIbcDMA with a value of 1 to enable DMA would return this
This error may also occur, if the board index passed to the ibdev function is incorrect. The default interface name for board is GPIB0, so it has a board index of 0, not 2.Solutions: Use the default settings for your GPIB hardware in the GPIB Configuration Utility Add Comments 1 2 3 4 5 My Profile|Privacy|Legal|Contact NI© National Instruments Corporation. Ni 488 Command Requires Gpib Controller To Be Controller In Charge EABO (6)Error Condition: I/O operation aborted.Description: EABO indicates that an I/O operation has been cancelled for some reason.Possible Cause: The EABO error is usually the result of a timeout during a
ESAC (5)Error Condition: GPIB board not System Controller as required.Description: ESAC results when ibsic, ibsre, SendIFC, or EnableRemote is called when the GPIB board does not have System Controller capability. This error occurs when a thread starts asynchronous I/O and then attempts to access the interface while the interface is still in an unsynchronized state. 11 No capability for operation. The other KnowledgeBase entries linked below offer more information on GPIB communication problems. YourFeedback!
Check your system's resource manager to see if another board is trying to use part or all of this address range. Gpib Tutorial All rights reserved. | ERROR The requested URL could not be retrieved The following error was encountered while trying to retrieve the URL: http://0.0.0.5/ Connection to 0.0.0.5 failed. Related Links: KnowledgeBase 2368N85R: ENEB Error (Non-existent GPIB Board Error) During Communication with InstrumentProduct Manuals: NI-488.2 User Manual for WindowsGPIB Support: Introduction to the Interactive Control (IBIC)Knowledge Base 1XOHEPPH:GPIB Error Codes Refer to the KnowledgeBase for more information about correcting errors in LabVIEW.
This error is returned when the operation cannot be performed because of the existing lock on the interface. The system returned: (22) Invalid argument The remote host or network may be down. Gpib Error Codes You should (almost) always leave your GPIB board configured for primary address 0 and no secondary address. How To Find Gpib Address The error can be due to an operating system error, such as when the operating system is unable to provide a physical address to the driver.
Check the user manual to see if your instrument needs to be in a GPIB or 488.2 mode in order to be a GPIB Listener. http://imagextension.com/error-code/ftp-return-code-10000-error-code-00009.php The message to the instrument may contain a command that the instrument does not understand. An ibwait call pending in the driver returns ERST in the following situations: Another thread in the same process calls ibonl using the same unit descriptor as ibwait Another thread or Make sure that at least two-thirds of your devices are powered on. Gpib Interactive Control
This error occurs when there is a problem with a table used by a driver function. Generated Mon, 17 Oct 2016 09:05:24 GMT by s_ac15 (squid/3.5.20) This often results when the system goes to and returns from a standby state.Solutions: Take all handles offline and reinitialize the application Quit the application and restart the system Disable standby my review here Generated Mon, 17 Oct 2016 09:05:24 GMT by s_ac15 (squid/3.5.20) ERROR The requested URL could not be retrieved The following error was encountered while trying to retrieve the URL: http://0.0.0.6/ Connection
The board index is the number portion of the interface name for the GPIB board, but many people make the incorrect assumption that it is the primary address of the board. Labview Error Code My Profile | RSS | Privacy | Legal | Contact NI © 2014 National Instruments Corporation. ECIC (1)Error Condition: Function requires the GPIB board to be the Controller-In-Charge (CIC).Description: Certain functions require your GPIB board to be the CIC - these functions are noted in the NI-488.2
The error also can be due to a hardware limitation, such as when the DMA controller cannot address your buffer and the driver cannot remap the buffer. 9 DMA hardware uP The system returned: (22) Invalid argument The remote host or network may be down. The system returned: (22) Invalid argument The remote host or network may be down. What Is Gpib Call ibpad 0 and ibsad 0 at the beginning of your program to properly configure your board’s address.
To send an EOS add it to the end of the string written to the instrument. See the related KnowledgeBase entry on "GPIB Error Codes and Common Solutions" (linked below) for a complete list of GPIB error codes and their common causes. This problem happens when the board is not physically plugged into the system, the I/O address specified during configuration does not mach the actual board setting, there is a system conflict get redirected here Possible Cause: The GPIB board is not configured to be the System Controller.
Answered Your Question? 1 2 3 4 5 Document needs work? EADR is also returned by the function ibgts when the shadow handshake feature is requested and the GPIB ATN line is already unasserted. Often instruments will only respond to the standard commands after they have been placed in this mode, and otherwise they will give an error despite being sent a valid command. Also make sure that processes do not lock an interface up for the entire period of execution.EARM (22)Error Condition: ibnotify callback failed to rearm.Possible Cause: This error occurs when we use
The GPIB address 0 is usually reserved for the controller (usually your National Instruments GPIB card in your computer). It is easy to misspell this message as "*IND?", which the instrument will not understand, so it will not generate a message string for you to read from the instrument. ENOL usually occurs when a write operation is attempted, but no Listeners are addressed or there are no Listeners at the specified address(es). Error on a WriteThe EABO (abort) status being returned from your GPIB write call (IBWRT function call) is the most common write problem.
Use the unit descriptor returned from ibdev or ibfind as the first parameter in subsequent NI-488 functions. Answered Your Question? 1 2 3 4 5 Document needs work? If your GPIB board is the System Controller, then use the ibsic function (or SendIFC function) to send an interface clear. Related Links: KnowledgeBase 1DREQIQA: GPIB Error Codes and Common Solutions (Part 1) KnowledgeBase 20H8G27Z: Why Can I Communicate with My GPIB Instrument in MAX, but Not in LabVIEW Product Manuals Library
Check the address of your instruments and make sure that they are different from the controller. An EABO error is usually caused by a TIMO (timeout) error where the controller card times out waiting for the handshake to complete. You may receive a timeout during write operations with a PCI-GPIB board, if the PCI bus mastering (an option in the BIOS of your computer) is not enabled. For example, I may install a PCI-GPIB board in my computer and give a primary address of 2.
Your cache administrator is webmaster. For example, a device-level ibwrt returns EBUS if it is unable to send the addressing command bytes. 15 Serial poll byte queue overflow. 16 SRQ stuck in ON position. Unable to Detect GPIB InstrumentThe Measurement and Automation Explorer (MAX) provides a "Scan for Instruments" function to quickly test your GPIB setup. Clicking on the "Scan for Instruments" button in MAX will conduct a FindLstn() test for all GPIB addresses 0-30 to find active listeners on the bus.
If your instrument is an older, non-IEEE 488.2 compliant device, then it will not understand "*IDN?", so it will not generate a message string for you to read from the instrument. The default timeout value is 10 seconds, so you could observe the write function wait for 10 seconds and then fail. ibgts also can return this error if the board is not properly addressed for shadow handshaking. 4 Invalid argument or arguments to function call.
© 2017 imagextension.com